The tinySA comes in a sturdy box for protection during transport
Inside the box you find the tinySA with some accessories
- two SMA cables and a barrel connector
- an antenna with an SMA connector
- a USB-C cable
After unpacking the tinySA insert the USB cable into the tinySA and connect the cable to a USB charger or computer USB port to charge the tinySA. One hour charging should be sufficient for first usage.
After charging the USB cable can remain connected or can be disconnected.
The next steps are also explained in the video on first use
Kurt Poulsen wrote a document on the tinySA that can be found through tinySA@groups.io
Use one of the supplied SMA cables to connect the low port to the high port.
Power the tinySA using the tiny power switch at the top.
Touch the screen to activate the menu system and select CONFIG and then SELF TEST. If navigation goes wrong use the BACK button or switch the tinySA Off and On to get back into a known state. If all goes well the self test will pass all tests and you can touch the screen once more to conclude.
Next step is to do the power calibration of the LOW INPUT MODE. THis has to be done only once for one frequency and it does not have to be repeated before every measurement as the level calibration is very stable and frequency independent.
Doing the level calibration for the high input mode is a bit more complex but clearly explained in this '''video'''
Now the tinySA is ready to do measurements.
WARNING!!!. The input signal must be below +10dBm otherwise the tinySA can be damaged.
You can always go back to a defined state by clicking the PRESET button in the input.