PRESET loads or saves configurations.
LOAD STARTUP reloads the startup presets.
LOAD X loads the settings from slot x
STORE supports updating the stored presets
STORE AS STARTUP sets the current settings as the startup preset.
STORE X stores the current settings in slot X.
FACTORY DEFAULTS resets the startup preset to the factory defaults.
FREQ sets everything related to the frequency range to scan.
START sets the scanning to start/stop mode and sets the start frequency
STOP sets the scanning to start/stop mode and sets the stop frequency
CENTER sets the scanning to center/span mode and sets the center frequency
SPAN sets the scanning to center/span mode and sets the frequency span
ZERO SPAN sets the scanning to center/span mode, sets the span to 0Hz and sets the center frequency
RBW sets the resolution bandwidth. Keep in mind a low RBW may increase scanning time substantially.
SPURREMOVAL activates the two spur removal mechanisms.
LEVEL sets everything related to the level of the signals being measured.
REF LEVEL sets the level in selected unit of the top of the display.
SCALE/DIV sets the amount of selected unit per division of the display.
ATTEN sets the amount of attenuation applied to the low input/output.
CALC selects various calculation options over time such as averaging or max hold.
UNIT selects the display unit. dBm, dBmV, dBuV, Volt or Watt
EXTERN AMP sets the level offset in dB caused by an external amplifier or attenuator.
TRIGGER selects the trigger mode
AUTO is the normal spectrum analyzer scanning mode, triggering is not active.
NORMAL will display a new scan as soon as a signal in the scan causes a trigger event.
SINGLE will wait for a signal to cause a trigger event and display the scan.
LEVEL activates a keypad for entering the trigger level.
UP EDGE generates a trigger event when the input level goes from below the set trigger level to above the set trigger level
DOWN EDGE generates a trigger event when the input level goes from above the set trigger level to below the set trigger level
DISPLA controls various aspects of how the signals are being displayed.
PAUSE SWEEP pauses the scanning
STORE TRACE writes the current measurement to the blue trace. Click again to store again.
CLEAR STORED removes the stored trace
SUBTRACT STORED subtracts the stored trace from the current measurement.
NORMALIZE will store the current scan and subtract this stored scan from future measurements.
WATERFALL displays the power level over time in a waterfall map.
SWEEPSETTINGS menu contains various settings on how to sweep the selected frequency or time span.
NORMAL sets the default sweeping mode
PRECISE sets the precise scanning mode.
FAST sets the fast sweeping mode.
SWEEP TIME sets the target time for a complete sweep in seconds.
SWEEP POINTS allows setting the number of sweep points.
FAST SPEEDUP allows setting the acceleration factor for fast sweeping.
MARJER controls the markers on the display.
MODIFY MARKER allows selecting a marker and shows a submenu to modify the [:../MARKER:TYPE].
MARKEROPS allows setting the frequency display range based on the active marker
SEARCH MARKER allows positioning a non-tracking marker on signal maxima or minima.
MEASURE helps to quickly set the TinySA for certain measurements.
OFF switches of any measurement related setting and behavior and returns the TinySA to regular operation
HARMONIC switches to a marker configuration for measuring the level of harmonics of a signal
OIP3 switches to a marker configuration for measuring the Output IP3 level of a signal
PHASE NOISE switches to a marker configuration for measuring phase noise of a signal
CONFIG activates the configuration menu.
TOUCH CAL enables the calibration of the touch panel. The results are stored in NVM.
TOUCH TEST is used to verify touch calibration.
SELF TEST is used after a cable is used to connect the low and high input/output.
LEVEL CAL is used to calibrate the power measurement levels.
VERSION shows the SW version information.
EXPERT CONFIG activates the expert configuration menu
LO OUTPUT enables the output of the first LO though the high connector
ACTUAL POWER allows calibration of the dBm reading by entering the know level of the signal under marker 1.
IF FREQ allows entering the IF frequency used in low mode.
SCAN SPEED allows setting the speed of scanning.
REPEATS sets the number of repeat measurements at each frequency.
MIXER DRIVE sets the power output to the mixer.
AGC enables/disables the build in Automatic Gain Control.
LNA enables/disables the build in Low Noise Amplifier.
BPF enables measurement of the performance of the internal Band Pass Filter.
BELOW IF switches the LO to below the IF when measuring below 190MHz.
HOLD TIME sets the number of scans after which a maximum measured with MAX DECAY will start to decay.
NOISE LEVEL sets the expected noise band in dB.
ACTUAL FREQUENCY sets the measured frequency of the 10MHz CAL OUTPUT
DFU switches the TinySA to firmware upgrade mode.
MODE activates the mode switching menu.